The doseBadge noise dosemeter can be hired for one week or more to assess the noise levels that your workers are being exposed to.
Particularly useful for people who move around a lot, the doseBadge kit contains five dosemeters, making it ideal for monitoring a larger number of people during the hire period.
Hire Kit Includes
Noise exposure in the workplace can be assessed using either a hand-held sound level meter or using a noise dosemeter mounted on the worker's shoulder.
With a hand-held sound level meter you measure the sound level at each location and use this with the length of time the workers spends there to calculate the exposure. With a noise dosemeter you measure for the full working shift and it provides you with the overall exposure value.
Each doseBadge (there are five in the kit) contains a rechargeable battery. The unit itself is small and light weight. It mounts on the worker's shoulder to measure the noise levels close to the ear.
The best way is to leave it measuring for the full working shift, providing you with the noise exposure level, LEPd, as required by the noise at work regulations.
The dBadge noise dosemeter is intended for measuring sound levels in a relatively noisy factory environment.
For sound level measurement in lower noise environments you should use a hand-held sound level meter. For longer term measurements these hand-held meters can be mounted on a tripod.
Leave the doseBadge measuring for the full working shift...
While it is possible to measure the LAeq over a shorter period and then calculate the exposure (LEpd), when using a dosemeter it is better to leave it measuring for the full working shift. This saves doing the potentially complex noise exposure calculations and ensures you measure all relevant noise levels.
The measurements can be viewed on the Reader's built in graphic screen or they can be transferred to a computer for storage, analysis and reporting. The Reader can download a number of doseBadges before the measurements are transferred to a computer, so it is only necessary to carry the Reader on site.
|Standards||IEC 61252:1993 Personal Sound Exposure Meters|
|ANSI S1.25:1991 Personal Noise Dosemeters|
|Reader's Internal Acoustic Calibrator to IEC 60942:2001 Class 2|
|Measurement Range||70 dB(A) to 130 dB(A) RMS|
|120 dB(C) to 140 dB(C) Peak|
|Measurement Functions||1 Minute Time History of: LAeq (3dB), LAVG (4dB or 5dB), Peak (C) Level, Battery Level|
|For 3dB Exchange Rate: LAeq, LEX,8h, LAE, % Dose, Exposure (Pa2h)|
|For 4dB & 5dB Exchange Rates: LAVG, TWA, % Dose, Estimated % Dose|
|Weightings||‘A ’ for all RMS measurements.|
‘C ’ for Peak Sound Pressure
|Environmental||Temperature: -10 °C to +50 °C Operating|
|Humidity: Up to 95%RH Non-Condensing|
|Dimensions||Microphone Apex Ø13.0mm,Base Ø47mm,Height 38mm|
|Weight||doseBadge: 45gms (1.6oz)|
Reader: 400gms (14oz)